Make New Entry,
Make Followup Entry
User name S.Abrahamyan
Log entry time 01:58:39 on July 09, 2010
Entry number 330943
This entry is a followup to: 330882
keyword=RHRS VDC problems
During short access to the Hall. I find out that flat cables lower V5 and
lower V11 were swapped at the input of level translators.
I changed them back, that happened after run 1906.
I made 2 database folders 20100708 and 20100709.
20100708 - reflects that cables were swapped.
20100709 - is normal database.
Runs 1900-1906 were taken on Jul 9, but before cables were fixed, so
these runs should be analyzed separately.
Cards lower V3 and lower V2 - noisy.
A copy of this log entry has been emailed to: email@example.com, firstname.lastname@example.org, email@example.com, firstname.lastname@example.org, email@example.com