Make New Entry,
Make Followup Entry
User name J. Zhang
Log entry time 15:06:53 on July10,2010
Entry number 331330
This entry is a followup to: 331318
keyword=Right Q1 was not recover to the previous after fixed
After replaced the power supply board in the morning, the right Q1 field was not recovered to the value (0.01414) before 06:30 July 10, 2010. The filed value was about 0.01402. When it tripped at 14:18, it was set to 0.01398 until we recognized the difference. Then we set it back to 0.01413 at 14:30. Runs between 2010 and 2020 may be affected by this issue.