While working on the argon manifold I found the leak was actually in the hall. The cap on the argon line for the RICH, on the Left Detector Stack, was only finger tight. I tightened the Swagelok cap and valved the line off at the argon manifold outside the Gas Shed. Some more clear labeling was put on the tubing. The argon gas consumption is now down to a more reasonable amount.