I pulled the FADC board from the Moller DAQ crate and put it on the shelf to be surveyed.
After that I'll take it to the electronics group to get the firmware upgraded. The new firmware changes the logic of the "accidentals" scaler from (CR.and.CL).and.(SR.and.SL delayed >100ns) to the more sensible (CR.and.SR).and.(CL.and.SL delayed >100ns).
The unit should be back in the crate by the end of the week.