The fuses on the Low voltage power supply for the calorimeter were replaced. All channels columns should be fine now. I switched back to a T5 trigger but did not get any triggers, I'll check with Magali if I have to initialize something to get it to work. The stop for the ARS is generated fine with the LED pulse arriving 80 ns after the LED trigger. I turned everything off for now.