The next part of the Moller commissionning was done. MCC had a problem with tunning the beam passing through the Compton chicane for Moller test. Finally we use straight beam for the Moller test. HV, pedestals and threshould tunnings for the Moller detector were done. Scans of the 1rs and the 3rd Moller quads were done to optimize Moller optics. The results will be posted on the Moller web page.