Make New Entry,
Make Followup Entry
User name hyde
Log entry time 00:46:57 on October16,2010
Entry number 335049
keyword=1024 Hz Clk Double Pulsing
1024 Hz clock (T8) was firing at 2044 Hz.
Problem was fanout to long cable (7 m) to DVCS DAQ
that wasn't terminated. This was installed approx 1 wk ago.
Cable is now terminated. No more double pulsing