Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 15:10:02 on February 6,2011
Entry number 346092
keyword=DAQ is working, use test17 and Hand
work done by R. Michaels, L. Selvy, V. Sulkosky, D. Anez
The DAQ is working now, with some qualifications.
startcoda Bbite .... and then select test17 (usually it will
be the choice since it was what ran last)
startcoda Hand .... and then select HANDalone.
(Bbite means BBite + L-HRS + R-HRS but no Hand).
(Hand means HAND alone)
1. The SFI Aux card on Hand corrupts the data in such a way to
cause SYNCH loss in EB. I don't want to explain the many hours
of isolating the problem to this card. Yes, it's the card.
We tried to find a spare card, but could not. I will hunt for it
Monday morning, perhaps Cuevas has one. After that we can put
Hand into the main DAQ.
2. Various things were wrong with the handshaking on L-HRS and
R-HRS, not worth documenting here. It works now. The next step is
to run with HRS singles and do the retiming. Then start checking
cosmics data on the two HRS. I suppose one could check cosmics
on Bigbite simultaneously ....
3. A crew of people are working on Hand as I write this ...