Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 15:36:15 on February 7,2011
Entry number 346122
keyword=Full DAQ running now
13:40 .... Replace bad SFI Aux card on ROC8 (Hand FB crate).
13:42 .... verify HANDalone works.
13:47 .... verify that test17 still works (from download).
13:54 .... routed the RS485 cable for Bigbite to include the
ROC8 crate (passively for now). Run test17. It works. This is
already better than yesterday because this was hanging up.
Fixed the TS11 hangup problem, it seems.
It was opening event type 100 (last scaler reading) but not
closing it, because of the hybrid way we were running. Anyway....
We restored "ts control" in the bb_sfi1.crl
(This means HANDalone will not work)
The THREEARM config runs now.
This is the full config with 11 DAQ components, including
Bigbite, Hand, L-HRS, and R-HRS.
We are adjusting the "startcoda" script so that it
starts the DAQ the way we need for the experiment.
Larry reminds me that the scripts have been taken out of
CODA because they were hanging up state transitions. I'll
try to put these back in later. I think it will work if
you invoke script1 which in turn invokes script2 and puts
it in the background .... or something.