Make New Entry,
Make Followup Entry
User name yez
Log entry time 11:59:08 on February 24, 2011
Entry number 348435
keyword=LHRS: move two cables of PRLs to other ADC channels
We have two cables which include several PRLs PTMs that give only
pedestals but not real signals. I tested both cables and it looks like
the cables themselves are good, and I also double checked the PMTs'
signals from Patch Panel downstairs and they are visible in the scope
which means that the PMTs are also OK. I decided to move the cable of
17L~32L from Slot24:16~31 to Slot25:16~31, and move the cable of 33,34L/R
from Slot25:16~31 to Slot25:32~47. The 33,34L/R was in 0~15 and David
moved it to 16~31, but the data taken after the change shows nothing
different as before so I decide to move it again, while put 17L~32L in
this group(25:16~31) to verify whether this group itself is bad or not.
I have a question is:
Is 1881ADC receiving positive ADC signals? Why I ask is because that I
found the 50ohm filters that connect from the ribbon cables and put into
the ADC module flip the signals from negative to positive.
A copy of this log entry has been emailed to: email@example.com,firstname.lastname@example.org,email@example.com