Make New Entry,
Make Followup Entry
User name A. Camsonne
Log entry time 17:44:26 on May 09, 2011
Entry number 358733
This entry is a followup to: 358720
keyword=Timing change understood
We looked at the signals with Zhihong and Vince.
We concluded that timing was made off the L1A instead of retimed signal.
After much bad cabling chasing we figured out it was coming from the
timing module which might not have been initialized properly.
Looking at the CODA config I found that TS0 field were blank, I copied
the object and initialization string from another CODA config which had
it and after resetting the configuration UBTHREEARM the timing was fine.
So timing should be fine now.