We took data for BPM calibration for 5 positions with both new and old constants and with 10 KHz receiver on and off. Pengjia can give a more detailed summary since I left in swing shift. One thing we noticed is that BPM position changes when at high current 8 uA pulsed and 100 nA so we have to be careful to take calibration data at the current we will run. A few runs for raster calibration were also taken. Signal on 1H05 is still very small and the noise got worse for 1H04 later in the night.