Make New Entry,
Make Followup Entry
User name Higinbotham
Log entry time 06:39:58 on March 11, 2012
Entry number 365611
This entry is a followup to: 365590
We are running 100 nA; just the calibration is off.
We fixed the blead thru issue last night by closing the Hall A slit.
Unless, someone at MCC opened it, it shouldn't be an issue.
We can run lower currents then 100 nA, though we were indeed worried
about Hall-C blead-through returning during the night.
A copy of this log entry has been emailed to: email@example.com, firstname.lastname@example.org, email@example.com, firstname.lastname@example.org, email@example.com