Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 14:21:50 on March17,2014
Entry number 386909
keyword=pulser runs for BPM pedestal, part 1
Runs starting at 1155 and 20192 are pulser runs.
I took over the DAQ in the counting room.
Signal values with no signal and "default" gain have been measured
in runs 1155 and 20193. They are typically 3500 - 4000 ADC channels.
Run 1156 and 20194 are with BPM 1H04A at forced gain = 0.
Leave 1H04B as is for now. (This is partly a check of cabling, too.)
Expected: BPM4A signals to shift and become very narrow but BPM4B
to remain wide.
Found: BPM signals on L-HRS remain same (wide) while all BPM signals
on R-HRS are narrow. Time to check cabling !