3He QuasiElastic/Online HowTo
From Hall A Wiki
Revision as of 18:09, 6 May 2009 by Jinhuang (New page: Here is shift index page of worker HowTo for experiment transversity. == General == * Data Replay * Experts On-Call == Detectors == * [[BigBite HV...)
Here is shift index page of worker HowTo for experiment transversity.
- BigBite HV GUI (updated 23 April 2009)
- High Voltage GUI for BigBite (Wire Chamber, Scintillator and Shower)
- Change Wire Chamber Threshold
- Recover after BigBite CAMAC reset
- How to set BigBite Total Shower thresholds
- BigBite Wire Chamber Trip Recover Procedure
- How to reboot TS12 ROC5-10
- RICH pedestal run
- Scaler problem or "Scaler problem: Please check the web page"
- How to calibrate shower and preshower with cosmics
- Neutron Detector