April 3rd 2023
From Hall A Wiki
Revision as of 10:52, 3 April 2023 by Camsonne (Created page with "*Updates **ECal ***No issue with ECAL, two layers went bad because of summing amplifier shower blocks- one module needs to still be repaired ***data taking looked fine ***Rpi...")
- No issue with ECAL, two layers went bad because of summing amplifier shower blocks- one module needs to still be repaired
- data taking looked fine
- Rpi issue for thresholds from time to time - readback
- beginning kin 4 not calibrated at beginning
- issues ?
- EPICS interface for HV CAEN could be improved
- add scaler information for hodoscope - FADC or hardware scaler for central bars
- RF signal : 2 1190 channels - 1 channel looks odd - need to be checked ? Maybe bad connection
- missing highest energy cluster TDC in HCAL
- channel missing - can recover TDC information from lower amplitude channel from cluster - but not ideal
- could still be software
- is HCAL TDC threshold to low ?
- need to check Sean for GEn status ?
- beamline : all looked ok - a few more crashes from sbsvme29 - could check if there are any data issues - will give information