- 1 The Hall A Compton
- 2 Shift Workers' Operation Guide
- 3 Compton Experts' Operation Guide
- 4 Compton Meetings
- 5 Current Compton Experts
- 6 Expert Information
- 7 Old Compton page (pre-Spring 2016)
The Hall A Compton
The Compton polarimeter is used to (non-invasively) monitor polarization of the incoming electron beam. It has three major components: an optics table with a laser that can lock to a Fabry-Perot cavity, an electron detector, and a photon detector to count scattered electrons and photons, respectively.
Last edit: LThorne 02:08, 24 February 2016 (UTC)
Shift Workers' Operation Guide
Start a Compton run every two hours.
The Compton DAQ is located in the back room in the Counting House. All commands are done from compton@compton.
If CODA is not already running:
- Kill CODA first, then start it up again:
$ kcoda $ startcoda
- A few new Terminal windows and a GUI pop up.
- Let "top" refer to the bar with named menu options near the top of the GUI (ex: Platform, Sessions, etc). Let "bottom" refer to the bar below "top" that contains a line of buttons (hover over them to reveal their labels; starting with "Configuration" on the left).
- In top, select: Platform > Connect
- In bottom, click on "Configuration" button.
- If configuration listed in the "Run Parameters" box is not FADC_prod: in top, select Configurations > Cool and change it to FADC_Prod.
- In bottom, click on "Download" button
- In bottom, click on "Prestart" button
- When ready to start run, click "Start" button. Typical run duration: 2 hours. Type in any Start Of Run comments in the box. Click "OK" (even if you left comments blank!).
- When ready to stop run, click "End" button. Type in any End Of Run comments in the box. Click "OK" (even if you left comments blank!).
Compton Experts' Operation Guide
Feb-March 2019 Commissioning
Procedure/Run plan for commissioning Compton Photon detector in Spring 2019 Compton Photon Detector Centering Procedure
- HV on/off? -> Compton#High Voltage Controls
- [Experts only]: Correct parameters for run in integratingDAQ.flags and compton.params
Run Compton analysis:
$ cd /home/compton/online/CompMon
Run analysis software:
$ ./compmon <run number, no brackets>
$ root -l /data/cmuwork/rootfiles/Fall2016/compmon_<run number, no brackets>.root
Run desired C script located in CompMon directory.
Some useful analysis scripts:
- /home/compton/lthorne/CompMon/plotAsymSpectrum.C: Quick and dirty view of asymmetry histograms, cavity and BCM limits.
- /home/compton/lthorne/CompMon/buildQuartets.C and buildQuartetsPlot.C: For constructing multi-run asymmetry plots.
- Run instructions are given in the comments at the beginning fo the file.
- /home/compton/lthorne/CompMon/BCMCal/calibration<runNumber>,C: BCM calibration scripts for specific runs.
- /home/compton/lthorne/CompMon/plotAccQuartets.C: plots sum, difference histograms for beam off, laser on, laser off for Acc0, Acc4.
- /home/compton/franklin/CompMon/plotPulser.C: peek at DAQ pulser channels, including info on 'variable' and 'delta' for channel 13.
Link to list of good Compton runs:
List of runs with comments, so know which ones were not junk.
Juan Carlos Google Docs: Runlist (Email Juan Carlos if you want to granted write permission)
Taking an LED run:
- Find parameter file GUI:
$ cd /home/compton/franklin/ $ ComptonConfig
- Click "Pulser On, Pulser Trigger" on parameter file. Click "Apply", then "Done" to close GUI.
- Go to CODA. Start a new run to apply the settings. (This does not need to be a long run. You may end it immediately.)
- Start your LED run.
- When you go back to normal Compton runs: don't forget to turn the pulser off and do a short run to update the parameter file settings!
- Primary Lead Collimator
- Presently left fully open at 20 mm diameter.
- This is a 10 cm long collimator with inserts which can reduce aperture down to 10 mm or 5 mm in diameter.
- Synchrotron Shields
- For PREX-II a single 0.25 mm lead foil (painted) is taped to the front of the photon detector.
- Compton Jaws
- Remote access for experts can be found at 
- Note, for the duration of the PREX-II running period, the Compton Tungsten Collimator System ("Compton Jaws") will not be used. It should be left it is's parked position of:
aperture = 1.5 mm y center = -46.0 mm
- Centering Beam & Photon Detector
- Refer to section [Compton Photon Detector Centering Procedure] for more info.
Photon Detector Table
For controls, troubleshooting: Compton Photon Detector Home position of photon detector table:
|Date||X [mm]||Y [mm]||Comment|
High Voltage Controls
- HV Control GUI
- GUI can only be accessed once per person. We share this GUI with Moller Polarimeter.
Software & Data Analysis
Relevant files, scripts, etc.
From within NewTools controls (see first few steps in Compton Photon Detector Compton GUI section):
- On far right of JTabs menu, there are "master" controls. Find and click on Tools Screen.
- Go to the helicity controls. (The default is delayed, so be careful of this.)
As a brief check, run plotAsymSpectrum.C.
See last list entry for most recent meeting.
- Meetings before 2014
- 2014 Meetings
- 2015 Meetings
- 2016 Meetings
- 2017 Meetings
- Monday, February 12, 2018, 11:00 am EST - Compton Electron Detector
- Monday, February 19, 2018, 11:00 am EST - Compton Electron Detector
- Thursday, May 10, 2018, 10:00 am EDT - Compton
- Thursday, May 24, 2018, 10:00 am EDT - Compton
- Thursday, June 14, 2018, 10:00 am EDT - Compton
- Thursday, July 5, 2018, 10:00 am EDT - Compton
- Thursday, July 19, 2018, 10:00 am EDT - Compton
- Thursday, August 9, 2018, 10:00 am EDT - Compton
- Thursday, August 23, 2018, 10:00 am EDT - Compton
- Thursday, September 6, 2018, 10:00 am EDT - Compton
- Wednesday, September 19, 2018, 11:00 am EDT - Compton
- Wednesday, October 17, 2018, 11:00 am EDT - Compton
- Wednesday, November 7, 2018, 11:00 am EST - Compton
- Wednesday, November 28, 2018, 11:00 am EST - Compton
- Wednesday, December 12, 2018, 11:00 am EST - Compton
- Wednesday, January 9, 2019, 11:00 am EST - Compton
- Wednesday, January 23, 2019, 11:00 am EST - Compton
- Wednesday, February 6, 2019, 11:00 am EST - Compton
- Wednesday, February 27, 2019, 11:30 am EST - Electron Detector
- Wednesday, March 20, 2019, 11:00 am EST
- Wednesday, April 17, 2019, 11:00 am EDT
- Wednesday, May 1, 2019, 11:00 am EDT
- Wednesday, May 8, 2019, 11:00 am EDT
- Wednesday, May 22, 2019, 11:00 am EDT
- Wednesday, May 29, 2019, 11:00 am EDT
- Wednesday, June 5, 2019, 11:00 am EDT
- Wednesday, June 12, 2019, 2:30 pm EDT
Current Compton Experts
|Juan Carlos Cornejo (Photon Detector)||x5282|
For expert monitoring and troubleshooting information please visit Compton Expert Information.
Code can be found on github: https://github.com/JeffersonLab/HallA_CompMon JC says:
The Makefile is ancient (since it has to work on old systems too) but there is an easy "fix". In the src/config.d directory, create a file similar to cornejo_compton_jlab.conf but modify the paths respectively. Then run ./config config.d/<your_config_file> It will update the silly Makefile accordingly
Some old data and examples can be found on
Old Compton page (pre-Spring 2016)
Last edit on 2019 06 24, by Cornejo