The High voltage for PMTs and GEMs for SBS experiments use at least 4 different types of power supplies. These include:
- 5 Lecroy mainframes
- A Caen 1527 mainframe
- A Wiener Mpod mainframe for (UVA) GEM detectors
- CAEN VME V6521 cards for (INFN) GEM detectors
Each of these systems has a vendor or lab developed graphical or web interface. In order to simplify operations, EPICS controls have been implemented for all these systems.
User control of high voltage
EPICS Graphical interfaces will implemented for all the HV under EPICS control. These interfaces will include the ability to backup and restore HV settings.
The various types of high voltage mainframes have vendor or user supplied control interfaces that don't use EPICS. These interfaces will continue to work.
The master copy of the EPICS HV controls for SBS are in the directory
~aslow/EPICS on adaqsc. The
EPICS directory contains the following
- The base EPICS distribution from https://epics.anl.gov/
- Contains CAEN libraries that are needed for the communicating with SY1527 HV crates
- Contains the source code for EPICS support to provide IOCs (Input Output Controllers) for all the models of HV crates in use. Also contains configuration files to define EPICS signals for all the HV channels in use.
The base-3.15.8 and sbs-epics directories are also installed on the Raspberry Pi for each Lecroy Mainframe and on sbsvme27 for controlling CAEN VME HV cards.
sbs-epics/apps contains these source code/configuration directories:
- Code and configuration for CAEN SY1527 mainframes. (Used by BigBite hodoscope)
- Code and configuration for CAEN VME HV cards. (Used by GEM detectors
- Scripts to start the IOCs for each of the HV systems
- Code and configuration for RPi enabled Lecroy Mainframtes
- 2.6-0App : A wrapper for running the EPICS IOCs
- Code and configuration for communicating with SNMP systems. Used for Wiener HV.
- Library routines needed by caenHvApp
This code is compiled by running make from sbs-epics/apps. The Makefiles in some of the above source code directories are setup to compile only under certain conditions:
caenHvApp and syX527App will only be compiled if the
clas12-epics-third-party-libs directory exists.
snmpApp will only be compiled if snmp support is installed on the
operating system. (By checking if
caenV65XXApp is only compiled if LINUXVME_INC is defined.
Every high voltage mainframe must be assigned a unique crate/mainframe number. For the Lecroy crates, this crate number is automaticallly assigned from the Raspberry Pi hostname. If the hostname is of the form rpiNN, then NN is the crate number. Other HV system types are assigned a crate number manually. The following mainframes are in use for the SBS eperiments:
0 CAEN1527 BigBite Hodoscope 8 Lecroy HRS Left 17 Lecroy BigBite 18 Lecroy BigBite 20 Lecroy Hcal 21 Lecroy Hcal 91 MPOD UVA GEM HV 92 CAEN V6521 INFN GEM HV
EPICS PV Naming Convention
The EPICS Process Variables (PVs) for individual HV channels are chosen to have a consistent form, independent of hardware. The form is chosen to match the built in EPICS on CAEN4527 mainframes. The use of consistent PV naming convention allows for GUIs that are independent of hardware.
EPICS PVs for HV have the form
"%s%d:%.2d:%.3d:%s", prefix, crate, slot, channel, property
where prefix is a string unique to Hall A ("HAHV"), crate is a mainframe number (from 0 to 99), slot is a card # within a mainframe (0-99), channel is the channel number within a card (0-999) and property is the attribute to be written or read. Properties common to all hardware types are:
V0Set (w) Voltage setting for a channel V0Setr (r) Readback of voltage setting VMon (r) Monitored/measured voltage I0Set (w) Setting for Current at which to trip channel I0Setr (r) Readback of trip current setting IMon (r) Measured current Pw (w) (0/1) Turn off or on the channel Pwr (r) Readback of on/off status RUp (w) Ramp rate in V/s when turning on channel Rupr (r) Readback of ramp up rate RDWn (w) Ramp down rate in V/s when turning off channel RDWnr (r) Readback of ramp up rate
These property names are chosen to be consistent with the property names of the CAEN4527 built in EPICS.
Building GUI screens
If detector HV is changed due to bad channels or othe reaons, control screens need to be rebuilt.
ProcServ and ProcServMgr
On systems where the EPICS IOCS (servers) run, a cron task is
configured to periodically run ProcServMgr. ProcServMgr looks at the
sbs-epics/apps/iocBoot/procServMgr.conf to determine which IOC
tasks need to run on that system and starts any IOCs that are not
running. The IOC tasks are run by ProcServ which allows experts to
login to the IOC with the command
telnet PORT where the
PORT number is defined in
Hardware specific notes
Hall A has a large number of Lecroy Mainframes. These were controlled either by ARCnet or RS232. Several years ago, the PC boards in each of these mainframes was replaced with Raspberry Pi boards that communcate with the mainframe via a custom serial board developed by Javier Gomez and Jack Segal. Two processes run on each board. A process that communicates with the HV cards (written by J. Gomez) and a process (shim) (written by Brad Sawatzky) that mimics the original ARCnet/RS232 command set, except that it uses TCP/IP. This is further described in HV HowTo for Experts.
In 2000, Chris Slominski developed an EPICS driver for the Lecroy mainframes that communicated with ARCnet. This driver has been adapted to use the TCP/IP interface that the modified Lecroy mainframes now provide.
A single linux IOC can be configured to support multiple mainframes or an IOC can be run on each rPI board. For the SBS HV, the later method is used. Each mainframe is configured to create create EPICS signals for the installed HV cards.
To install EPICS on a Lecroy Mainframe, use the following steps.
- Setup a rpi with a hostname of rpiN, where N is not an existing HV crate number (See "Crate Numbering" section). (Unless it is replacing an existing crate.)
- Insure that the rPI board in the mainframe has the software prerequisites to compile C programs (gcc, ncurses, readline) and the perl modules needed by the shim (IO-Multiplex, Net-Server, IO-Tty, Expect). If the JAVA GUI already works with this mainframe, then these prerequisites are installed. If they are not, ask the Spectrometer Support Group to prepare an RPI with these prerequisites.
- Obtain the files scripts.tar.gz, lecroy_servers.tar.gz and epics.tar.gz and place them in the home directory of the pi account on the rPI. These can be found in the directory
- Untar all three files into the pi account home directory.
~/scripts/kill_hv_allto kill and disable old versions of the Lecroy software.
- Wait 5 minutes and check that an "i2lchv" process and a "LecroyHV_shim" procees is running.
- Check that the EPICS server is running with
telnet localhoset 20000. You should see the
dblshould show a long list of EPICS signal names starting with "HAHV91".
Lecroy Cron job
The cronjob that is setup with
crontab ~/scripts/cron.tab run the
start_hv_cron_new every four minutes. This script checks
that each of the servers, i2lchv, the shim and the epics server is
running. It will start any server that is not running.
If the EPICS server is not running,
start_hv_cron_new will first run
~/scripts/configure_epics.py. This python script does several
- Determines the mainframe/crate number from the hostname rpiN.
- Reads the file
/home/pi/scripts/LecroySlotList.txtto find which # slots are occupied and what type of module is in each slot. Currenly, only 1461N, 1461P and 1471N cards are supported.
- Using the slot information, define epics signals for every HV # channel in the crate by writing the file
The CAEN1527 is used for the BigBite hodoscope. It is instrumented with four 48 channel A1932 cards. These cards differ from most other HV cards. There is a single primary power supply and 48 secondary channels which supply the PMTs. The voltage on the primary supply can be set to an arbitrary value. The secondary channels can only be set to the primary voltage or less. Not all attributes are availabe on both primary and secondary channels. When an attribute doesn't exist, the GUI will show -1.0.
UVA GEM modules. Wiener MPOD systems are controlled by SNMP (Simple Network Management Protocol). The standard EPICS [https://groups.nscl.msu.edu/controls/files/devSnmp.html SNMP EPICS driver] is used.
The IOC for the Wiener MPOD system runs on adaqsc. It is started by ProcServMgr.
CAEN VME V6521 cards
CAEN VME V6521 cards are used for the INFN GEM detectors. These cards
do not use a custom HV mainframe, just a regular VME crate. This
EPICS driver for these cards runs on the Linux Single board computer
(sbsvme27) in the VME crate with the cards. The driver software for
these cards is in
sbs-epics/apps/caenV65XXApp and the startup script for the IOC
sbs-epics/apps/iocBoot/iocv65xx/st.cmd. Slot number used in the EPICS PVs is
not the physical slot but rather the logical slot, where slot 0 is the
first card defined, slot 1 is the second card defined etc. The cards
are defined with the command
V65XXInit(Ncards, baseaddr, incaddr)
st.cmd. For exmple,
0x100000) will define "slots" 0-2 with VME addresses
0xa00000, 0xb00000, and 0xc00000.