Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 23:21:15 on August22,2009
Entry number 285221
This entry is a followup to: 285209
keyword=DAQ restored, ROC2 replaced.
I swapped out the ROC2 cpu with a 5100 board that has 500 MByte
of memory. Looks like it works ok.
Also, a non-existent scaler was being addressed in the SFI,
an artefact of recent FADC work. I removed those lines from
the CRL file.
I did not make any other change. In particular the FADC remains
in the SFI but is not addressed.