HV HowTo for Users
High Voltage Instructions -- the basics
The High Voltage is controlled by LeCroy 1458 crates and associated modules like the 1461N. A trailing "N" means negative, and a "P" means positive HV.
Simplified Instructions : To control the HV in Hall A, do the following
Login as "aslow" on the "adaqsc" computers.
To run the Left HRS alone: ./hvs LEFT
To run the Right HRS alone: ./hvs RIGHT
To run the beamline crate: ./hvs BEAMLINE
Links to more HV help
These two links provide some old but useful information.
TroubleShooting for Users
After installing the rpi boards as servers, it seems that less goes wrong. If you have a setup that doesn't use rpi, you should upgrade.
1. Restarting the server sometimes is necssary. See the expert page https://hallaweb.jlab.org/wiki/index.php/HV_HowTo_for_Experts. Simplest way to restart the server is to power-cycle the rpi and wait 5 minutes. The server gets restarted by "cron".
2. Is the HV crate on ? There are two power supplies which must be turned on. Turning "on" in software will only turn on the HV if the hardware switches are in the right state and the electrical cables are plugged in. Sorry, I know it's obvious, but I've seen this mistake ...
3. If multiple cards seem "bad", I've found that power-cycling the HV crate helps. And what I mean by power-cycling is you go to the crate and manually turn it off (cutting power), then on. NOT just via the GUI but actually the hardware. And there is both a mainframe to power-cycle and the 24-Volt supply. Plus the rpi has a source of power. Altogether 3 things that can be power-cycled (mainframe, 24V, rpi).
4. Need the sign ! Most of our cards are negative HV, and when you enter the target HV, you must also enter a minus sign. Like "-852" and not just "852". Beware !
5. Make sure that HV cards are all well-seated in the sockets, and no stress on the cables that can unseat the cards. Also, it has been observed before that a "bad" card can cause inexplicable problems. Finding and and replacing the bad card solves this. One way to find a bad card is to observe what the lecroy servers print when run by hand. If one card is persistently missing, it's probably bad. But this might be too "expert" an instruction.