Second floor counting house DVCS - CSR test setup
Port 3 DVCS MVME 6100 halladaq1
5 Motor Controller
7 VME crate
To access the port type :
telnet hatsv15 200[port]
From adaq computer : ssh -Y adaq@adaql6
telnet hatsv15 2003
V265 0x0A A24
Lecroy 1182 0x0099
Test SIS FADC 0x08
Lecroy 1151N A24/A16 0x2222
V1190 0x0020 ( top to bottom for MSB ) 0
V792 from SRC 0x0A83
I will setup the V1495 as a LVDS to NIM converter.
I put it in the Flash ADC crate for now.
Adress in A32 is 0x0900 ( modifier is 0x9 )
Crate name is hallavme8
In A32, the local adress is : 0x81000000 this version works only in A32
I can see the board register at :
Software from Roman to load the user content is located in a-onl account in v1495 directory
Mezannine D has NIM Mezannine E has ECL output Mezannine F has NIM
-- x_IDCODE : -- 000 : A395A (32 x IN LVDS/ECL) -- 001 : A395B (32 x OUT LVDS) -- 010 : A395C (32 x OUT ECL) -- 011 : A395D (8 x IN/OUT NIM/TTL) -- Expansion Mezzanine Port Signal Standard Select -- x_LEV : -- 0=>TTL,1=>NIM
-- Expansion Mezzanine Port Direction -- x_DIR : -- 0=>OUT,1=>IN
2 additionnal V1495
0x0220 This board is set in A24 Adress is 0x0220 ( modifier is 0x39 )
this will return the corresponding address which is for the VME6100 0x90200000
New VXS FADC test setup for DVCS/NPS December 4th 2014
VME CPU was put on the network vmedvcs2 22.214.171.124
I created a new database dvcs and session DVCS to be used for CODA on aonl1.
startcoda script was modified to start CODA on adaq account on aonl1.